Hantek MSO5202D 200MHz, 2/16 Channel Mixed Signal Oscilloscope
Hantek MSO5202D 200MHz Mixed Signal Oscilloscope
Mixed Signal Oscilloscope; 200MHz Bandwidth; 1GSa/s Real Time Sample Rate;1MRecord Length; Large (7.0-inch) color display,WVGA(800x480); 16 channels Logic Analyzer,500MSa/s sample rate; Power Trigger mode: edge/pulse width/line selectable video/slop/overtime etc.
- 16 channels logic analyzer + 2 channels oscilloscope + external trigger.
- Big and clear display (7.0-inch color LCD, high revolution 800 x 480), clear lifelike waveform display.
- 1GSa/s real time sampling rate.
- USB host, support flash memory card storage and USB interface system upgrade.
- Ultrathin design, handy volume, easily portable.
- VGA Optional
- Bandwidth 200MHz.
- Each channel record length up to 1M.
- Real time sampling rate up to 1GSa/s.
- Powerful trigger function.
- 32 kinds of automatic measurement function.
Logic Analyzer Functions
- 16 channels divided into 2 groups which is able to setup threshold level individually.
- Real time sampling rate up to 500MSa/s.
- Powerful trigger function: edge, duration, pulse width, code-type, queen, repeat.
|Sample Rate||Real-Time Sample: 1GS/s|
|Normal||Normal data only|
|Peak Detect||High-frequency and randon glith capture|
|Average||Wavefom Average, selectable 4,8,16,32,64,128|
|Inputs Coupling||AC, DC, GND|
|Inputs Impendance||1MΩ±2% ‖20pF±3pF|
|Probe Attenuation||1X, 10X|
|Supported Probe Attenuation Factor||1X, 10X, 100X, 1000X|
|Maximum Input Voltage||CAT I and CAT II: 300VRMS (10×), Installation Category;
CAT III: 150VRMS (1×);
Installation Category II: derate at 20dB/decade above 100kHz to 13V peak AC at 3MHz* and above. For non-sinusoidal waveforms, peak value must be less than 450V. Excursion above 300V should be of less than 100ms duration. RMS signal level including all DC components removed through AC coupling must be limited to 300V. If these values are exceeded, damage to the oscilloscope may occur.
|Sample Rate Range||500MS/s--1GS/s|
|Waveform Interpolation||(sin x)/x|
|SEC/DIV Range||2ns/div to 40s/div, in a 2, 4, 8 sequence,|
|Sample Rate and
Delay Time Accuracy
|±50ppm?at over any ≥1ms time interval)|
|Position Range||2ns/div to 10ns/div;
(-4div x s/div) to 20ms;
|Delta Time Measurement Accuracy
|Single-shot, Normal mode:± (1 sample interval +100ppm × reading + 0.6ns);
>16 averages:± (1 sample interval + 100ppm × reading + 0.4ns);
Sample interval = s/div ÷ 200
|Vertical Resolution||8-bit resolution, all channel sampled simultaneously|
|Position Range||2mV/div to 10V/div|
|Rise Time at BNC( typical)||1.8ns|
|Analog Bandwidth in
Normal and Average
modes at BNC or with probe, DC Coupled
|2mV/div to 20mV/div, ±400mV; 50mV/div to 200mV/div, ±2V
500mV/div to 2V/div, ±40V; 5V/div, ±50V
|Math||+, -, *, /, FFT|
1024 sample point
|Low Frequency Response (-3db)||≤10Hz at BNC|
|DC Gain Accuracy||±3% for Normal or Average acquisition mode, 10V/div to 10mV/div;
±4% for Normal or Average acquisition mode, 5mV/div to 2mV/div
|DC Measurement Accuracy,
Average Acquisition Mode
|When vertical displacement is zero, and N ≥16:± (3% × reading + 0.1div + 1mV) only 10mV/div or greater is selected;
When vertical displacement is not zero, and N≥16: ± [3% × (reading + vertical position) + 1% of vertical position + 0.2div]; Add 2mV for settings from 2mV/div to 200mV/div; add 50mV for settings from 200mV/div to 10V/div
|Volts Measurement Repeatability,
Average Acquisition Mode
|Delta volts between any two averages of ≥16 waveforms acquired under same setup and ambient conditions|
|Trigger Types||Edge, Video, Pulse, Slope, Over time, Alternative|
|Trigger Source||CH1, CH2, EXT, EXT/5, AC Line|
|Trigger Modes||Auto, Normal, Single|
|Coupling Type||DC, AC, Noise Reject, HF Reject, LF Reject|
(Edge Trigger Type)
1div from DC to 10MHz;1.5div from 10MHz to 100MHz; 2div from 100MHz to Full;
DC(EXT): 200mV from DC to 100MHz; 350mV from 100MHz to 200MHz;
DC(EXT/5): 1V from DC to 100MHz;1.75V from 100MHz to 200MHz;
AC: Attenuates signals below 10Hz;
HF Reject: Attenuates signals above 80kHz;
LF Reject: Same as the DC-coupled limits for frequencies above 150kHz; attenuates signals below 150kHz
|Trigger Level Range||CH1/CH2: ±8 divisions from center of screen;
|Trigger Level Accuracy( typical)Accuracy is for signals having rise and fall times ≥20ns||CH1/CH2: 0.2div × volts/div within ±4 divisions from center of screen;
EXT: ± (6% of setting + 40mV);
EXT/5: ± (6% of setting + 200mV);
|Set Level to 50%(typical)||Operates with input signals ≥50Hz|
|Video Trigger Type||CH1, CH2: Peak-to-peak amplitude of 2 divisions;
|Signal Formats and Field Rates, Video Trigger Type||Supports NTSC, PAL and SECAM broadcast systems for any field or any line|
|Holdoff Range||100ns ~ 10s|
|Pulse Width Trigger|
|Pulse Width Trigger Mode||Trigger when (< , >, = , or ≠); Positive pulse or Negative pulse|
|Pulse Width Trigger Point||Equal: The oscilloscope triggers when the trailing edge of the pulse crosses the trigger level.
Not Equal: If the pulse is narrower than the specified width, the trigger point is the trailing edge. Otherwise, the oscilloscope triggers when a pulse continues longer than the time specified as the Pulse Width.
Less than: The trigger point is the trailing edge.
Greater than (also called overtime trigger): The oscilloscope triggers when a pulse continues longer than the time specified as the Pulse Width
|Pulse Width Range||20ns ~ 10s|
|Slope Trigger Mode||Trigger when (< , > , = , or ≠ ); Positive slope or Negative slope|
|Slope Trigger Point||Equal: The oscilloscope triggers when the waveform slope is equal to the set slope.
Not Equal: The oscilloscope triggers when the waveform slope is not equal to the set slope.
Less than: The oscilloscope triggers when the waveform slope is less than the set slope.
Greater than: The oscilloscope triggers when the waveform slope is greater than the set slope.
|Time Range||20ns ~ 10s|
|Over Time Mode||Rising edge or Falling edge|
|Time Range||20ns ~ 10s|
|Trigger on CH1||Internal Trigger: Edge, Pulse Width, Video, Slope|
|Trigger on CH2||Internal Trigger: Edge, Pulse Width, Video, Slope|
|Trigger Frequency Counter|
|Readout Resolution||6 digits|
|Accuracy (typical)||±30ppm (including all frequency reference errors and ±1 count errors)|
|Frequency Range||AC coupled, from 4Hz minimum to rated bandwidth|
|Signal Source||Pulse Width or Edge Trigger modes: all available trigger sources
The Frequency Counter measures trigger source at all times, including when the oscilloscope acquisition pauses due to changes in the run status, or acquisition of a single shot event has completed.
Pulse Width Trigger mode: The oscilloscope counts pulses of significant magnitude inside the 1s measurement window that qualify as triggerable events, such as narrow pulses in a PWM pulse train if set to < mode and the width is set to a relatively small time.
Edge Trigger mode: The oscilloscope counts all edges of sufficient magnitude and correct polarity.
Video Trigger mode: The Frequency Counter does not work.
|Cursor Measurement||Voltage difference between cursors: ?V
Time difference between cursors: ?T
Reciprocal of ?T in Hertz (1/ΔT)
|Auto Measuerment||Frequency, Period, Mean, Pk-Pk, Cycli RMS, Minimum, Maximum, Rise time, Fall Time, +Pulse Width, -Pulse Width, Delay1-2Rise, Delay1-2Fall, +Duty, -Duty, Vbase, Vtop, Vmid, Vamp, Overshoot, Preshoot, Preiod Mean, Preiod RMS, FOVShoot, RPREShoot, BWIDTH, FRF, FFR, LRR, LRF, LFR, LFF|
|Logic Analyzer Specifications|
|Max. Input Impendence||200K(C=10p)|
|Input Voltage Range||-60V~60V|
|Logic Threshold Range||-8V~8V|
|Max. Sample Rate||500MHz|
|Compatible Input||TTL, CMOS, ECL|
|Edge Trigger||D0-D15 select slope (rising or falling edge)|
|Pulse Width||D0-D15 select pulse polarity (positive or negative pulse), trigger when (=, ≠, >, <), trigger pulse width|
|Code-type||D0-D15 select code-type (H, L, X)|
|Duration||D0-D15 select persist time and trigger when (data terminate, data start, and data delay)|
|Queue||D0-D15 select specific data index (0-3) and code-type (H, L, X)|
|Repeat||D0-D15 select code-type (H, L, X) and repeat times|
|Display Type||7 inch 64K color TFT (diagonal liquid crystal)|
|Display Resolution||800 horizontal by 480 vertical pixels|
|Display Contrast||Adjustable (16 gears) with the progress bar|
|Probe Compensator Output|
|Output Voltage( typical)||About 5Vpp into ≥1MΩ load|
|Supply Voltage||100-120VACRMS(±10%), 45Hz to 440Hz, CAT?
120-240VACRMS(±10%), 45Hz to 66Hz, CAT?
|Fuse||2A, T rating, 250V|
|Temperature||Operating: 32? to 122? (0? to 50?);
Nonoperating: -40? to 159.8? (-40? to +71?)
|Humidity||+104? or below (+40? or below): ≤90% relative humidity;
106? to 122? (+41? to 50?): ≤60% relative humidity
|Altitude||Operating: Below 3,000m (10,000 feet);
Nonoperaring: Below 15,000m(50,000 feet)
|Size & Wight|
|Size||385mm x 200mm x 245mm|
†Some products must be specially ordered, usually resulting in a 1–2 business day delay. Please contact us for an estimated delivery date.